A blog on the repair, operation and calibration of surface analysis systems and components including electron spectrometers, sputter ion guns and vacuum related hardware. Click on the Index tab below to see a list of all posts. Visit our website at http://www.rbdinstruments.com
RBD Instruments’ microCMA is an affordable compact Auger Electron Spectroscopy (AES) cylindrical mirror analyzer package which is small enough to add to existing vacuum chambers. For applications which require elemental information on the top most mono layers of specimens, AES can be the perfect solution. More information…
Following are links to a variety of Auger Electron Spectroscopy tutorials which explain in detail the principles of AES, the instrumentation, and the applications.
A new feature has been added to AugerScan – the option to automatically mark all the peaks of a particular Auger element when marking the primary peak. With this feature you can mark an element’s primary peak and AugerScan will automatically mark the rest of the element’s peaks and select the primary for atomic concentration.
An Example Using Copper (Cu1)
AugerScan – Marking multiple peaks for copper (Cu1)
In the example above, we’ve selected the carbon (C1), oxygen (O1) and copper (Cu1) peaks. AugerScan automatically marked the additional copper peaks (Cu2, Cu3, and Cu4). When performing an atomic concentration calculation, only the Cu1 peak is selected.
You can of course easily change which peaks are selected for atomic concentration by clicking the “Markers” command from the “Edit” menu. You can also remove selected markers from peaks using this dialog.
AugerScan – Marker Selection Dialog
To turn this feature on/off, choose the “Options…” command from the “Data” menu, and select/deselect the checkbox labeled “When Marking and Element, Mark all of its Peaks”.
AugerScan – Marking all Peaks Option
(This feature was originally developed for CMapp, RBD’s software for the microCMA Compact Auger Analyzer)
The model 32-100 electron multiplier supply is used on older Physical Electronics Auger electron spectroscopy and X-ray photoelectron spectroscopy surface analysis systems to control the electron multiplier voltage.
When using the 32-100 electron multiplier supply in the digital mode (the software automatically sets the electron multiplier voltage) the auto-ems box in the Auger Scan software AES electron multiplier properties dialog box needs to be checked, and the CMA multiplier switch on the front panel of the 32-100 must be set to digital.
But what if the 32-100 still does not work in the digital mode? In that case, there may be a problem with one of the digital ICs. The following procedure explains how to try and repair this problem.
Turn off the 32-100 and remove the cover.
Most 32-100s have three ICs for the SED digital side of the control which are not used. You can move those ICs over to the CMA side of the control and see if that solves the problem.
Remove ICs U2, U4 and U6. These are the CMA digital ICs. Then, move over U3, U5 and U7 from the SED side if those chips are available. If they are missing, then you will need to order some of those ICs from Digikey, Newark or RBD Instruments. U2 and U4 are 74LS174s and U6 is an AD7521.
If you did have those ICs and that did not solve the problem, then it most likely means that one of the encoder circuit ICs are defective. Those are U16 (74123) and U1 (AM25LS2538). RBD Instruments provides these parts.
You do not need to replace the SED digital ICs as they are not used.
It is assumed that the 32-100 is working in the analog mode. If the 32-100 is not working in the analog mode it will not work in the digital mode either.
Refer to the pictures below for the locations of the ICs on the 32-100 motherboard.
Note that the switch positions need to be set as shown below.
If you need further help troubleshooting your 32-100 electron multiplier supply please contact RBD Instruments dot com